Hot Air Oven- Principle, Parts, Types, Uses, Examples
A hot air oven is a device that uses dry heat to sterilize various items and materials. It was originally developed by Louis Pasteur, a French microbiologist, and chemist, in the 19th century. A hot air oven can operate at high temperatures ranging from 50 to 300 degrees Celsius and can effectively kill microorganisms such as bacteria, viruses, fungi, and their spores by destroying their cellular components and proteins.
Dry heat sterilization is a physical method of sterilization that involves exposing the items to be sterilized to hot and dry air for a specific period of time. It is suitable for materials that can withstand high temperatures and do not get damaged by oxidation, such as glassware, metal instruments, powders, oils, and waxes. Dry heat sterilization has some advantages over other methods of sterilization, such as:
- It does not require water or steam, which reduces the risk of corrosion, rusting, or contamination of the items.
- It is economical and easy to operate, as it only requires electricity and a thermostat to control the temperature.
- It can achieve higher temperatures and faster sterilization than moist heat methods, such as autoclaving or boiling.
- It leaves no toxic residues or waste chemicals on the items, unlike some chemical methods of sterilization.
However, dry heat sterilization also has some limitations and disadvantages, such as:
- It is not effective against some heat-resistant microorganisms, such as prions, which are proteinaceous infectious agents that cause neurodegenerative diseases.
- It is not suitable for materials that have low melting points or are sensitive to heat, such as rubber, plastic, paper, or cotton.
- It is time-consuming and requires longer exposure times than moist heat methods to achieve complete sterilization.
- It may cause damage or alteration to some materials due to oxidation or thermal expansion.
Therefore, dry heat sterilization by hot air oven should be used cautiously and only for appropriate materials that can tolerate high temperatures and dryness. Moreover, the efficacy of the sterilization process should be verified by using appropriate indicators, such as temperature-sensitive tapes or biological indicators using bacterial spores.
A hot air oven works on the principle of dry heat sterilization, which involves killing microorganisms by exposing them to high temperatures for a prolonged period of time. The dry heat causes dehydration, oxidation, and denaturation of cellular components, leading to the death of the microorganisms.
The hot air oven consists of a heating element that generates heat and a fan that circulates the hot air inside the chamber. The temperature and time of sterilization depend on the type and volume of the material to be sterilized, as well as the type and number of microorganisms present. The most commonly used settings are 160°C for 60 minutes or 170°C for 30 minutes.
The hot air oven can sterilize items that are heat-resistant and do not melt or catch fire at high temperatures, such as glassware, metal instruments, powders, oils, etc. However, it is not suitable for items that contain moisture, such as rubber, plastics, surgical dressings, etc., as they may be damaged by the dry heat. Moreover, some microorganisms, such as prions and endospores, may be resistant to dry heat and require moist heat or other methods of sterilization.
A hot air oven comprises various parts that play a crucial role in its functioning. The major parts are:
- Insulated Chamber: To sustain the desired temperature inside, its insulated chamber is crafted from premium-grade materials like stainless steel. It has a rectangular shape and space for ribs to keep shelves at different levels. The chamber can vary in size and capacity depending on the model and intended use oven.
- Shelves: These are objects that hold the items to be sterilized and are made of aluminum or stainless steel. Their number may differ depending on the quantity in addition to the weight and size of the item`s capacity of the oven. They are placed on the ribs with some gaps to facilitate the movement of air. Some shelves may have openings for aeration.
- Door: There is a single door on one side installed. Heavy hinges and a handle for opening and closing. The door has an asbestos gasket on the side to reduce heat loss during operation. Some models may have a double-glass observation window to allow the user to monitor the heating process inside the chamber.
- Heating Elements: These are devices that generate heat by passing an electric current through a conductor. They are usually located at the bottom or sides of the chamber and can operate at temperatures from 50 to 300 degrees Celsius. There are different types of heating elements used in hot air ovens, such as circular, U-shaped, wave-shaped, square-shaped, three-sided, or four-sided heaters.
- Fan: This is a device that is driven by a motor and is used to distribute hot air evenly inside the chamber. It ensures that the items are always in contact with hot air and dry faster. Some models may have more than one fan for better air circulation.
- Thermostat: This is a device that senses and regulates the temperature inside the chamber. It is connected directly to the heating elements and can resist extreme heat with a high negative temperature coefficient. It allows the user to set and maintain the desired temperature in the hot air oven and avoid temperature overshoot**: This is a device that displays the internal temperature of the oven. It can be either a thermometer or a thermocouple that is attached to the chamber wall or door. Some models may have digital displays for more accuracy and convenience.
- Timer: This is a device that controls the duration of the heating or sterilization process. It can be either electrical or mechanical and can operate for 5 to 60 minutes, depending on the setting. Some models may have alarms or indicators to notify the user when the time is up.
- Fuse: This is a device that protects the oven from electrical damage due to high currents during short circuits or overloads. It is usually located in the control panel or near the power supply cord and breaks the circuit when it detects excessive current flow.
- Control Panel: This is the region that allows the user to adjust and monitor different parameters such as temperature, time, power, etc. It also has indicator lamps (usually green for power and red for the heater) and switches knobs for turning on and off the oven.
These are some of the main parts of a hot air oven. However, there may be variations in design and features depending on the manufacturer and model of the oven. Therefore, it is advisable to read the user manual carefully before operating a hot air oven.
Hot air ovens can be classified into different types based on the mechanism of air circulation and distribution within the chamber. The main types of hot air ovens are:
- Natural convection oven/gravity convection oven: It works on the principle of natural convection, in which heated air at the bottom of the device rises to the ceiling, begins to cool, and then descends to the floor, allowing the cycle of heating and cooling to continue naturally within the device. It has a non-uniform temperature distribution and is suitable for powder samples.
- Oven with forced convection: The forced convection oven uses a motorized fan/blower to force hot air to distribute uniformly throughout the chamber`s interior. Because the sample is constantly in contact with hot air, it absorbs heat and dries faster. It has better temperature uniformity and stability than natural convection ovens.
- Side draught oven: It has a unidirectional flow of air from one side to the other, allowing samples to absorb maximum heat in a short period of time. It is ideal for preheating plastic clothes in hospitals, as well as smooth sheets or plates.
Some other types of hot air ovens are:
- Vacuum oven: It uses a vacuum pump to create a low-pressure environment inside the chamber, which reduces the boiling point of liquids and speeds up the drying process. It is suitable for heat-sensitive materials or samples that contain solvents.
- Cleanroom oven: It is designed to meet the stringent requirements of cleanroom applications, such as pharmaceutical, biotechnology, or semiconductor industries. It has a HEPA filter to remove airborne contaminants and a stainless steel construction to prevent corrosion and contamination.
- Laboratory oven: It is a general-purpose oven that can be used for various applications, such as baking, curing, annealing, sterilizing, drying, etc. It has a wide range of temperature settings and can accommodate different sizes and shapes of samples.
Before using a hot air oven, it is important to follow some safety precautions and guidelines to ensure proper functioning and avoid any damage or injury. The following are the steps for using a hot-air oven:
- The oven should be placed on a stable and flat surface, away from any flammable or combustible materials. The power cord should be plugged into a suitable electrical outlet with a proper voltage and frequency.
- The oven should be. Before placing the items on the trays or shelves, heat them for 30 minutes. This will ensure a uniform temperature distribution inside the chamber and reduce the sterilization time.
- The items to be sterilized should be wrapped in paper or aluminum foil and placed in metal or cardboard containers. The containers should have some openings or vents to allow the hot air to circulate around the items. The items should not touch each other or the walls of the chamber, and there should be enough space between the trays or shelves for airflow.
- The temperature gauge should be set at the desired level, depending on the type and volume of the items to be sterilized. The most commonly used temperature and time for sterilization are 170°C for 30 minutes, 160°C for 60 minutes, and 150°C for 150 minutes. However, these values may vary depending on the manufacturer`s instructions and the nature of the microorganisms being targeted.
- The door should be closed tightly by fastening the screws or latches provided, after which the temperature will start to rise. The power indicator light will turn on, indicating that the oven is working. The temperature indicator will show the current temperature inside the chamber, which should reach the set value after some time.
- The timer should be set according to the required holding time for sterilization. The timer will start counting down once the set temperature is reached. The heating indicator light will turn on and off periodically, indicating that the heating elements are cycling to maintain the temperature.
- Once the timer reaches zero, an alarm will sound, indicating that the sterilization process is complete. The oven should be turned off by pressing the power switch or button. The oven should be allowed to cool down before opening the door, to prevent thermal shock and glassware breakage.
- The items should be removed carefully from the oven using thermal gloves or tongs. They should be placed on a clean and dry surface and allowed to cool down completely before use or storage. The items should not be touched with bare hands or contaminated with any other materials until they are ready for use.
- The oven should be cleaned and maintained regularly according to the manufacturer`s instructions. Any spills or residues should be wiped off with a damp cloth and mild detergent. The trays or shelves should be washed with soap and water and dried thoroughly. The oven should be checked for any signs of wear or damage and repaired if necessary.
Hot air ovens have the following applications and use:
- It is used to sterilize laboratory equipment such as glassware (flasks, pipettes, Petri-plates, and test tubes), culture media, metal items (forceps, spatulas, scalpels, and scissors), non-volatile compounds (zinc and starch powder, sulfonamide), and other oil-containing materials. It can also sterilize heat-resistant injectable and surgical dressing materials.
- It can be used to test food items, pharmaceutical products, and other consumable materials for temperature stability during shelf life.
- It can be used in research settings for various experiments requiring controlled heating conditions in biology, chemistry, and material science.
- It can be used for heat treatment and drying of samples, such as metals, alloys, soil, and other materials. It can also be used for baking, curing, and aging polymers, rubber, and plastics.
- It can be used for heating various materials such as wax, oil, and food products.
A hot air oven is a widely used device for dry heat sterilization and other applications. However, like any other instrument, it has its own advantages and limitations. Here are some of them:
Benefits of a hot air oven
- Unlike an autoclave, it does not require water for sterilization. This makes it suitable for materials that are sensitive to moisture or steam.
- It is economical and easy to operate. It has a simple design and does not need complex maintenance or calibration.
- It can function at higher temperatures and faster than an autoclave. It can achieve temperatures up to 300°C and sterilize items in less than an hour.
- The small size of the oven requires less space and has an easier installation process. It can be placed on a benchtop or a shelf without much hassle.
- The dry heat does not corrode or rust metals or other sharp articles. It preserves the quality and durability of the items.
- It is non-toxic. No harmful chemical residues will be discarded after the sterilization process. It is also environmentally friendly as it does not produce any waste or emissions.
- The dry heat can penetrate deeply into thick objects such that it helps to achieve an in-depth sterilization effect. It can kill microorganisms that are resistant to moist heat, such as spores and prions.
Limitations of hot air oven
- It is not suitable to sterilize items such as rubber, plastics, surgical dressing, etc., due to their low melting point. They may deform or melt under high temperatures and dry heat.
- It may not destroy the heat-resistant endospores and prions completely because they utilize dry heat instead of moist heat. Some microorganisms may survive the sterilization process and cause contamination or infection.
- It is time-consuming relative to steam, flaming, chemical sterilization, or radiation. It takes longer to reach the desired temperature and to cool down after the operation. It also requires a longer holding time for effective sterilization.
- It may cause damage to some materials due to thermal expansion and contraction. The items may crack or break due to the rapid change in temperature. They may also lose their color or texture due to oxidation or dehydration.
A hot air oven is a device that uses dry heat to sterilize various items and materials. However, it also poses some potential hazards and risks if not used properly. Therefore, it is important to take some precautions when using a hot-air oven to ensure safety and efficiency. Here are some of the precautions that you should follow:
- Before using the hot air oven, make sure that it is clean and free of dust, dirt, grease, or any other contaminants that may affect its performance or damage its parts.
- Check the electrical connections and wiring of the hot air oven and make sure that they are in good condition and properly grounded. Avoid using extension cords or power strips that may overload the circuit or cause fire hazards.
- Place the hot air oven on a stable and level surface away from flammable materials, liquids, or sources of heat or moisture. Ensure that there is enough ventilation and clearance around the oven to prevent overheating or accumulation of gases.
- Do not use the hot air oven for purposes other than sterilization or heating. Do not place any combustible, explosive, volatile, or corrosive substances in the oven, as they may cause fire, explosion, or damage to the oven or the operator.
- Only use materials and items that are compatible with dry heat sterilization in the hot air oven. Avoid using plastics, rubber, paper, cloth, or other materials that may melt, burn, or decompose at high temperatures.
- Wrap the items to be sterilized in aluminum foil or place them in metal containers with lids. This will prevent contamination and ensure uniform heat distribution. Do not overfill the containers or stack them too closely, as this may hinder air circulation and affect sterilization efficiency.
- Set the temperature and time according to the type and volume of the items to be sterilized. Do not exceed the recommended temperature and time, as this may damage the items or cause overheating of the oven. Use a thermometer or a thermocouple to monitor the temperature inside the oven and adjust it as needed.
- Do not open the door of the oven during the sterilization process, as this may cause heat loss and temperature fluctuations. Wait until the sterilization cycle is completed and the oven has cooled down before opening the door. Use gloves or tongs to remove the items from the oven and avoid touching any hot surfaces.
- After using the hot air oven, turn it off and unplug it from the power source. Let it cool down completely before cleaning it with a damp cloth or a mild detergent. Do not use abrasive materials or solvents that may scratch or corrode the oven. Store it in a dry and clean place, away from direct sunlight or moisture.
By following these precautions, you can use the hot air oven safely and effectively for your sterilization needs.
There are many models and brands of hot air ovens in the market, each with different features, specifications, and prices. Here are some examples of hot air ovens that you can find online or in stores:
- Heating Oven EG (Steelco SpA): This is a high-performance oven that can be customized according to the customer`s needs. It has a single or double-door pass-through design, a chamber volume ranging from 330 to 10,000 liters, and a temperature range of 50 °C – 160 °C. It also has a self-washing system for easy cleaning and maintenance.
- Heat Oven FAC-100 (Tech-Lab Scientific Sdn Bhd): This is a compact and economical oven that has a microprocessor-PID controller for precise temperature regulation. It has a double-walled chamber with fiberglass insulation, adjustable stainless-steel shelves, and a capacity of 100L. It can operate at a temperature range of 5 °C – 220 °C.
- Sterilization Oven FN series (Nüve): This is a sterilization oven that uses natural air convection for uniform heat distribution. It has a triple insulation system that reduces energy consumption and heat loss. It also has a door that can be pressed tightly on the chamber gasket to prevent temperature loss. It can be used for dry heat sterilization in medical, veterinary, dentistry, and research settings.
- Heating Oven WHL, WHLL series (Tianjin Taisite): This is a heating oven that has a unidirectional airflow from one side to the other for fast and efficient heating. It has removable shelves, an air duct lateral plate, stainless steel inner chamber, and a bottom heater covering. It also has a double-glass observation window, silicon rubber seal, and air-tight buckle lock on the door. It has a temperature controller that protects from over-temperature and displays the temperature with a dual digital tube.
- Heating Oven 101A series/ 202A series (Zhejiang FUXIA Equipment Co., Ltd): This is a small-sized oven that can fit in small spaces. It has a glass observation window that allows the user to watch the heating process. It also has an inner chamber made of high-quality cold-rolled sheets that are stable, rigid, strong, earthquake-resistant, and corrosion-resistant.
These are just some examples of hot air ovens that you can choose from depending on your needs and budget. You can also compare their features, specifications, and prices online or in stores before making a purchase.
We are Compiling this Section. Thanks for your understanding.